The ANAIS-112 experiment at the Canfranc Underground Laboratory
The ANAIS experiment aims at the confirmation of the DAMA/LIBRA signal at the Canfranc Underground Laboratory (LSC). Several 12.5 kg NaI(Tl) modules produced by Alpha Spectra Inc. have been operated there during the last years in various set-ups; an outstanding light collection at the level of 15 photoelectrons per keV, which allows triggering at 1 keV of visible energy, has been measured for all of them and a complete characterization of their background has been achieved. In the first months of 2017, the full ANAIS-112 set-up consisting of nine Alpha Spectra detectors with a total mass of 112.5 kg was commissioned at LSC and the first dark matter run started in August, 2017. Here, the latest results on the detectors performance and measured background from the commissioning run will be presented and the sensitivity prospects of the ANAIS-112 experiment will be discussed.
The ANAIS (Annual modulation with NaI(Tl) Scintillators) experiment
intends to confirm the DAMA/LIBRA modulation signal
 using the same target and technique in a different
environment, at the Canfranc Underground Laboratory (LSC,
Laboratorio Subterráneo de Canfranc) in Spain. This goal imposes
strong experimental requirements: an energy threshold at or below
2 Experimental set-up
The nine modules used in ANAIS-112 were produced by AS in Colorado and then shipped to Spain along several years, arriving at LSC the first of them at the end of 2012 and the last by March, 2017 (see table 1). Each crystal is cylindrical (4.75” diameter and 11.75” length), with a mass of 12.5 kg. NaI(Tl) crystals were grown from selected ultrapure NaI powder and housed in OFE (Oxygen Free Electronic) copper; the encapsulation has a mylar window allowing low energy calibration. Two Hamamatsu R12669SEL2 photomultipliers (PMTs) were coupled through quartz windows to each crystal at LSC clean room. All PMTs have been screened for radiopurity using germanium detectors in Canfranc. The shielding for the experiment consists of 10 cm of archaeological lead, 20 cm of low activity lead, 40 cm of neutron moderator, an anti-radon box (to be continuously flushed with radon-free air) and an active muon veto system made up of plastic scintillators designed to cover top and sides of the whole ANAIS set-up (see figure 2). The hut housing the experiment is at the hall B of LSC under 2450 m.w.e..
|\brDetector||Quality powder||Arrival date||Light collection||K activity||Pb activity|
|\mrD0||90 ppb K||December 2012||15.31.1||1.1||3.15|
|D1||90 ppb K||December 2012||14.80.5 *||1.4||3.15|
DAQ hardware and software of ANAIS-112 were tested in previous ANAIS set-ups. For each module, individual PMT charge output signals are digitized and fully processed. Triggering is done by the coincidence (logical AND) of the two PMT signals of any detector at photoelectron level in a 200 ns window, enabling digitization and conversion of the two signals. There is redundant energy conversion by QDC modules and the building of the spectra is done off-line by adding the signals from both PMTs. The muon detection system based on plastic scintillators is fully implemented, allowing to tag muon-related events and to monitor on-site the muon flux. The slow control system is also operative, monitoring different parameters like radon activity, humidity, pressure, several temperatures, N flux or PMT High Voltage. In addition, a blank module will be set-up to monitor non-NaI(Tl) scintillation events and build a “blank” population for the study of annual modulation systematics.
3 Detector performance
The light output measured for all AS modules is at the level of 15 phe/keV, which is a factor of two larger than that determined for the best DAMA/LIBRA detectors . The fourth column of table 1 shows the preliminary results for the total number of photoelectron per keV using ANAIS-112 data, following the same method applied in ANAIS-25 and ANAIS-37 set-ups, described in . The new estimate is in very good agreement with the previous ones for D0-D5 detectors. This high light collection, possible thanks to the excellent crystal quality and the use of high quantum efficiency PMTs, has a direct impact in energy threshold. Triggering below 1 keV is confirmed by the identification of bulk Na and K events at 0.9 and 3.2 keV, respectively, thanks to coincidences with the corresponding high energy photons following the electron capture decays to excited levels.
Effective filtering protocols for rejecting non-bulk scintillation events, similar to those described at  and optimized for each detector, have been applied. Multiparametric cuts based on the number of peaks in the pulses, the temporal parameters of the pulses and the asymmetry in light sharing between PMTs are considered. Acceptance efficiency curves from external calibration data are obtained for each detector.
4 Radiopurity and background
Detailed background models for the first modules operated in ANAIS-25 and ANAIS-37 set-ups were developed , based on Geant4 Monte Carlo simulations and an accurate quantification of background sources: the intrinsic crystal activity directly assessed, the cosmogenic activity in crystals (precisely quantified from ANAIS-25 data ) and the activity from external components measured with HPGe detectors in Canfranc. At the region of interest, crystal bulk contamination is the dominant background source. Contributions from K and Na peaks and the continua from Pb and the considered cosmogenic H are the most relevant ones.
The ANAIS-112 data taken up to July, 2017 in the commissioning run have been analyzed to make a first quantification of the relevant background sources. The activity of K and Pb in the nine NaI(Tl) crystals has been determined and preliminary results are reported in the fifth and sixth columns of table 1. As made in previous set-ups, the potassium content has been deduced by identifying coincidences between the 3.2 keV emissions and the 1460.8 keV gamma-ray following the electron capture decay of K ; the obtained values are compatible with estimates from previous set-ups when available. Some detectors have similar content to that of DAMA/LIBRA crystals ; the average K activity in ANAIS-112, although higher than that of DAMA/LIBRA, is more than one order of magnitude lower than in large, low-background crystals tested from other suppliers. The activity of Th and U in the crystals is quantified by the measured alpha rates, following Pulse Shape Analysis (to distinguish alpha interactions from beta/gamma ones) and analysis of BiPo sequences; it is at a level of a few Bq/kg, but Pb out of equilibrium has been observed for all the modules. The origin of a possible Pb contamination was under study in collaboration with AS, which allowed to obtain lower activity in the last produced crystals (see table 1 and figure 2).
Figure 3 presents the background spectra at the low energy region in ANAIS-112 commissioning run; first and latest data are compared, showing the decay of cosmogenics in the last detectors. Preliminary background models for those modules (see one example in figure 5), considering the measured crystal activities and the ANAIS-112 configuration, point to equivalent relevant background sources in the very low energy region. The Pb contribution around 50 keV (see figure 3) is consistent with the measured alpha specific activity in all cases.
5 Sensitivity prospects
The prospects of ANAIS-112 for the identification of an annual modulation signal have been evaluated in  in terms of the a priori critical and detection limits of the experiment. The analysis is based on the detector response and the background level measured for the first modules operated in Canfranc. In particular, an average background (corrected for the cut efficiency) has been estimated in the regions of interest and five years of data taking have been assumed. Considering the variance of the estimator of the modulated amplitude, it is shown  that ANAIS-112 in 2-6 keV has a detection limit for a model-independent annual modulation (not related to a dark matter origin) below the measured amplitude by DAMA/LIBRA . As it can be seen in figure 5 (taken from ), under the dark matter hypothesis, for a detection limit at 90% C.L. and a critical limit at 90% C.L., ANAIS-112 can detect the annual modulation in the 3 region compatible with the DAMA/LIBRA result.
A few days after TAUP2017, Professor J.A. Villar passed away. Deeply in sorrow, we all thank his dedicated work and kindness. This work has been financially supported by the Spanish Ministerio de Economía y Competitividad and the European Regional Development Fund (MINECO-FEDER) under grants No. FPA2011-23749 and FPA2014-55986-P, the Consolider-Ingenio 2010 Programme under grants MULTIDARK CSD2009-00064 and CPAN CSD2007-00042 and the Gobierno de Aragón and the European Social Fund (Group in Nuclear and Astroparticle Physics). We acknowledge technical support from LSC and GIFNA staff.
- Attending speaker.
- Present Address: Centro de Investigaciones Energéticas, Medioambientales y Tecnológicas, CIEMAT, 28040, Madrid, Spain.
- Present Address: Università di Roma La Sapienza, Piazzale Aldo Moro 5, 00185 Rome, Italy.
- Electron equivalent energy.
- Bernabei R et al 2013 Eur. Phys. J. C 73 2648
- Sarsa ML et al 1997 Phys. Rev. D 56 1856
- Cuesta C et al 2014 Eur. Phys. J. C 74 3150
- Amaré J et al 2014 Nucl.Instrum. Meth. A 742 197
- Amaré J et al 2016 J. Phys. Conf. Ser. 718 042052
- Bernabei R et al 2008 Nucl. Instrum. Meth. A 592 297
- Oliván MA et al 2017 Astropart. Phys. 93 86
- Amaré J et al 2016 Eur. Phys. J. C 76 429
- Amaré J et al 2015 JCAP 02 046
- Cuesta C et al 2014 Int. J. Mod. Phys. A 29 1443010
- Coarasa I et al 2017 Annual modulation of dark matter: The ANAIS112 case Preprint arXiv:1704.06861v1 [astro-ph.IM]